|Determination of Paeoniflorin by HPLC
a. Methanol (HPLC) b. Nanopure water
HPLC (Shimadzu) Analytical balance (1/10000) Ultrasonic cleaner
0.45μm PTFE filter 50ml volumetric flask Screw cap vials
3) HPLC condition
Column: Kromasil C18 5m 4.6×200mm
Detector: UV 230nm
Mobile Phase: A=Water B= Methanol
Flow Rate: 1.0ml/min
Accurately weigh 50mg extraction into 50ml volumetric flask. Add Methanol to volume and sonicate to mix, filter through a 0.45μm PTFE filter into a screw cap vial and proceed with the analysis.
Standard preparation /calibration
Prepare a standard solution (approximately 0.1mg/ml in water) using a commercial reference standard of Paeoniflorin，Sonicate for about 10 minutes and transfer to screw-cap vials and can be stored in the refrigeration.
Determine the peak areas of the Paeoniflorin on the sample chromatogram and compare to the corresponding peaks in the chromatogram for the reference standard solution. Calculate the weight percent.